Parts Page 23 Page 24 – Appendix I: Do NOT adjust all 4 feet. Corner Test Section 6: The breakdown of the zones are as follows: Specifications Page 13 – Environment Page 14 – Section 6: Scale instrument, modular designed instrument, configurable and upgradable using printed circuit modules pages.
|Date Added:||21 October 2010|
|File Size:||65.8 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
To print the manual completely, please, download it. Press the [ZERO] key to zero the scale.
Fairbanks Ultegra SCB-R9000-14U Service Manual
Page 17 Section 6: Turn the unit upside down and remove all four feet from the scale platform 3. Calibration Procedures -8, -7, -6, -5, -4, -3, -2, -1, 0, 1, 2, 3, 4, 5, 6, 7, 8, 9, 10 The breakdown of the zones are as follows: This manual also for: Remove the scale from the packing box and place on a flat surface where it will be used.
Verify all cables within the scale are down flush and will not contact the platform.
Page of 29 Go. Scale instrument, modular designed instrument, configurable and upgradable using printed circuit modules pages. Page 15 Section 6: Customer Care Page 11 Page 12 – Section 5: A low power USB device draws less than mA. If the corners are not within one 1 division of each other, conduct a good visual inspection to verify that there are no binds or interference with the platform and re test the scale.
Ultegra Flat Top (SCB-R)
The scale may be placed on a desk, bench, or used with an ‘Insert’ adapter to countersink into a counter-top. Page 16 The scale must be calibrated if you proceed past this step.
Page 2 Every effort has been made to provide complete and accurate information in fajrbanks manual. Scale will display “”, when power is turned on and weight is present. If the corners are still not within one 1 division of each other, identify the corner s with the error and replace the load cell in that corner. Every effort has been made to provide complete and accurate information in this manual.
The maximum USB cable length is 5 meters unless a self powered hub is used to extend the cable length.
Using the level bubble, adjust one or two feet minimally to level the platform. By pressing ‘print’ button you will print only current page. Verify no weight is on the platform.
Weight readings will be accurate as soon as the unit is powered on and set to zero. Using the Scale 3. Comments to this Manuals Your Name.
Ultegra Roller Top (SCB-R)
With the platform empty, press the [ZERO] key. It uses default Windows drivers and is installed automatically upon first connection with a PC.
Parts Page 23 Page 24 – Appendix I: Disconnect power to the scale 2. Don’t show me this message again. Don’t have an account?
Page 23 Section 8: When the unit is powered, the liquid crystal display LCD will show the software part number and the revision followed by either “0. Corner Test Section 6: All equipment should be protected from direct sunlight.